Payment & Shipping Terms:
|Product Name:||PLC Melt Flow Rate Meter MFR & MVR||Die Length:||8.000±0.025mm|
|Cylinder Length:||152±0.1 Mm||Cylinder Diameter:||Ф9.550±0.025mm|
|Length Of Piston Head:||6.350±0.100mm||Diameter Of Piston Head:||9.475±0.015 Mm|
|Power:||220V±10% 60Hz||Temperature Constant Accuracy:||±0.2℃|
rubber tensile testing machine,
rubber hardness tester
ASTM D3364 PLC Control Melt Flow Rate Meter MFR & MVR For PC , FI , Plastics etc
ZL-3014C Melt Flow Indexer meets the requirement of GB/T3682-2000,ASTM D3364(sand ASTM D1238-98.And also ,we design and manufacture the instrument after we made a reference of similar standards ,such as JB/T5456 and ISO 1133 .It is used to determine the melt mass-flow rate (MFR) and the melt volume-flow rate (MVR) of thermoplastics materials .
It adopts auto sampling method and the cut samples are weighed up by balance when it is used to determine MFR .When it is used to determine MVR ,it measures the time which the piston is used to move the specified distance first through the displacement transducer and auto timer system and calculates MVR according to the formula later.
The final results of the two methods are all displayed on the machine’s LCD screen and printed and exported by computer .The instrument has high accuracy on data measurement with simple operation and stable capability.
It is not only suitable for the tests of engineering plastics with higher melt temperature ,such as PC ,FI plastics, nylon ,but also for the plastics with lower melt temperature ,such as PE ,PP ,POM ,PS ,ABS resin and poly carbonic acid ester ,etc .
It has wide range of applications in some industries ,such as plastic fabrication ,plastic goods ,petrochemical industry ;collage and university and science and research institute and commodity inspection department .
1. Extrusion section :
|Cylinder length||152±0.1 mm|
|Length of piston head||6.350±0.100mm|
|Diameter of piston head||9.475±0.015 mm|
2.Temperature range : 50℃-450℃
3. Temperature constant accuracy : ±0.2℃
4. Resolution :0.1℃
5. Power :220V±10% 60Hz
6. Standard test load (kg ,8 grades)
1st grade :0.325kg = (piston stem +weight tray +heat insulation sheath +weight # 1) = 3.187N
2nd grade :1.200kg = (0.325+weight # 2 0.875) = 11.77N
3rd grade :2.160kg = (0.325+weight # 3 1.835) = 21.18N
4th grade :3.800kg = ((0.325+weight #4 3.475) =37.26N
5th grade :5.000kg = (0.325+weight #5 4.675) = 49.03N
6th grade :10.000kg = (0.325+weight #5 4.675+weight #6 5.000) =98.07N
7th grade :12.500kg = (0.325+ weight #5 4.675+weight #6 5.000+ weight #7 2.500) =122.58N
8th grade :21.600kg = (0.325+ weight #2 0.875+ weight #3 1.835+ weight #4 3.475+ weight #5 4.675+weight #6 5.000+ weight #7 2.500+weight #8 2.915) =211.82N
The test load relative error shall not be more than ±0.5% .
Contact Person: Emily