Payment & Shipping Terms:
|Product Name:||Temperature Thermal Infrared Sensor Sterilizer Gate With High Sensitivity Probe||Temperature Measurement Range:||-20℃~100℃|
|Temperature Measurement Error:||±0.1℃||Traffic Rate:||1.5 Seconds Per Person (average Data)|
|Operating Voltage:||3.3~5V, Operating Current: 5mA||Working Power Supply:||Ac110-220v, 50/60hz|
food metal detector,
conveyor metal detector
Temperature Thermal Infrared Sensor Sterilizer Gate With High Sensitivity Probe
It is suitable for crowded places such as railway station, hospital, school, bank, office building, military compound, office hall of each institution, residential area, large factory and vegetable market.
The sterilizer for temperature measuring doorpost adopts high-precision infrared temperature sensor with the induction range of 0-5cm.
High precision and resolution, immunity to the environment and sunlight, small interference, room temperature measurement error ±0.1℃.The initial temperature is set at 37.3℃ (adjustable), and the detection door will alarm if the detected body temperature exceeds 37.3℃.After the pedestrian temperature measurement, you can reach down into the disinfection machine, automatic induction drops out of the wash disinfectant, so as to achieve the disinfection effect, to avoid hand virus constantly everywhere infection.
|Temperature measurement range||-20℃~100℃|
|Temperature measurement error||±0.1℃|
|Traffic rate||1.5 seconds per person (average data)|
|Operating voltage||3.3~5V, operating current: 5mA|
|Working power supply||ac110-220v, 50/60hz|
|Disinfectant||standard 1L, for 1000 people/time|
|Power||﹤ 20 w|
|Net weight||30 kg|
|Overall size||1850mm (height) ×150mm (width) ×330mm (depth)|
|Material||304 stainless steel|
Blue screen: normothermia Yellow screen: low fever warning Red screen: high fever alarm
Below 37.3℃ 37.4℃-37.9℃ 38℃or Higher
The installation of this equipment is very simple. It is recommended to choose an environmental area where the room temperature does not exceed 30 degrees and the low temperature does not fall below -10 degrees;
After selecting the installation area of the device, fix the device at the bottom and plug it in and use it.
Note: Please choose a level ground to install the equipment to prevent the equipment from tilting; the bottom must be fixed firmly to avoid falling down.
1. Power on: Plug in the power supply (this product is powered by AC 220V), touch the power button on the monitor for about 6 seconds, and after you hear the "Didi Didi" four beeps, the display is completed.
2. Start the temperature measurement: When the device is turned on, or the device is in the standby state, you need to start the temperature measurement. You only need to use the hand or forehead to irradiate the temperature in the temperature measurement area. When you hear a "drop", the display turns on. The temperature measurement function has been started.
3. Detection: Pedestrians enter the detection range and bring their foreheads close to the detection area (5-20CM). When they hear a "drop" from the device, the detection is completed and takes 1-2 seconds; the detection results will appear immediately On the display.
4. Disinfection: This equipment is equipped with an automatic induction disinfection machine. When the pedestrian detection result is normal body temperature, the pedestrian puts his hand under the automatic induction disinfection machine, the hands from bottom to top, the automatic induction disinfection machine drips the disinfection solution, and the pedestrian You can leave after rubbing your hands.
5. Standby: When the device does not perform temperature measurement for more than 60 seconds, the display of the device will go out, and the device's temperature measurement function will enter the standby state.
Matters needing attention:
1. Do not measure for a while or immediately after breastfeeding
2. Do not drink, eat or exercise before or during the measurement
3. If there is dust, hair or sweat on the temperature measuring probe, please wipe it with alcohol or disinfecting water immediately
4. Don't leave the temperature measurement area when you don't hear the "drop" at the end of the measurement
6. Please try to measure in the specified measurement area
Warranty and after sales:
(1) The following conditions are not within the scope of free maintenance. We can provide paid services. Please note:
1. Product damage caused by failure to install or use according to the instruction manual
2. The product exceeds the warranty period
3. The use of non-our company's accessories caused damage to the machine
4. The product has been disassembled or repaired by personnel not authorized by our company
5. Product damage caused by unexpected factors or human behavior, such as mechanical damage, drop, severe oxidation of the product, etc
6. Damage caused by transportation, loading and unloading during customer's return
7. Product damage caused by other irresistible natural forces such as earthquakes, fires, etc.
8. Failure or damage caused by other issues than the design, technology, manufacturing, quality, etc. of the product itself
(2), Special instructions
1. If there is obvious man-made hard object damage, cracks, severe deformation, etc., it will not be replaced. Users can purchase it separately.
2. The warranty replacement is limited to the core parts, and the accessories such as packaging and various technical materials are not covered by the warranty replacement. If the equipment performance problem occurs within 15 days after the product is purchased, and the appearance is not scratched, you can directly replace the new product. Warranty replacement products must be tested by our company. If it is a good opportunity, it will be returned directly to the customer. If the fault is confirmed, the same model product will be replaced, and the defective spare parts replaced will belong to our company.
(3) These regulations shall become effective on the date of promulgation.
(4) This product is guaranteed for one year from the date of sale and shipment
Contact Person: Wendy