Payment & Shipping Terms:
|Product Name:||Single - Column Helmet Wearing Device Strength Tester With PLC Touch Screen||Timing Range:||0-99min, Accuracy 0.1s.|
|Displacement Range:||0-500mm, Accuracy 0.1mm||Force Range:||0～2000N, Accuracy 1%.|
|Beam Speed:||0-500mm/min When No Load, Stepless Speed Change Can Be Adjusted. It Is Convenient To Quickly Adjust The Position According To The Helmet.||Type A And Type B:||During The Test, Type A 113N/min, Type B 75N/min, Etc. Are Automatically Switched According To The Specific Conditions Of The Test.|
|Artificial Chin:||Rigid Shaft Diameter 25mm, Two Shaft Spacing 75mm, Shaft Length 50mm.||Power Supply:||AC220V, 50Hz.|
|Standard:||GB811-2010||Key Words:||Helmet Tester|
AC220V Helmet Testing Equipment,
PLC Touch Screen Helmet Testing Equipment
1. The main machine structure is single-column structure, which can move up and down in a straight line.
2. Extra-large stretch stroke of the lace, stretching up to 500mm, overcomes the disadvantage of inaccurate deformation caused by the force of the lower stroke of the lace slipping from the adjusting buckle.
3. The initial load can be set by itself, and the deformation is automatically collected after the initial load, without manual resetting.
4. Touch screen + PLC + special test program control during test. The program automatic control speed does not need to be manually adjusted, the program is automatically adjusted according to the test situation, and the test data and results are automatically judged and processed.
5. Test hold The test program is automatically closed-loop controlled, and the constant force value of 30S or 120S is automatically maintained. There is no need to manually adjust according to the change of the force value.
6. Real-time display of force value, time, displacement, deformation, speed, test curve, etc. during the test.
According to the motorcycle helmet GB811-2010, it is mainly used to test the strength performance of the wearing device of the motorcycle helmet. It is widely used in related production, scientific research, quality supervision and testing units of motorcycle helmets.
According to the regulations of GB811-2010 motorcycle helmets, the strength performance of the helmet wearing device should meet
1. There shall be no elongation exceeding 25mm, tearing of the lace, falling off of the connector and loosening of the buckle.
|Force range||0～2000N, accuracy 1%.|
|Displacement range||0-500mm, accuracy 0.1mm|
|Timing range||0-99min, accuracy 0.1s.|
|Artificial chin||Rigid shaft diameter 25mm, two shaft spacing 75mm, shaft length 50mm.|
0-500mm/min when no load, stepless speed change can be adjusted. It is convenient to quickly adjust the position according to the helmet.
|Type A and Type B||During the test, Type A 113N/min, Type B 75N/min, etc. are automatically switched according to the specific conditions of the test.|
|Power supply||AC220V, 50Hz.|
Contact Person: Fiona