Payment & Shipping Terms:
|Product Name:||PLC Mask Partical Filtration Efficiency Tester ( PFE ) With PSL Test Aerosol||Test Area:||100cm²|
|Counter Sampling Flow Rate:||2.83L / Min ± 5%||Counter Laser Light Source:||Laser Diode|
|Screen Display:||Color 7-inch Touch Screen||Data Export Interface:||USB|
|Power Supply:||AC 220V / 50Hz||Meet Standard::||GB 19082, GB / T 32610, YY 0969, ISO-29463|
abrasion resistance testing machine,
yarn twist tester
PLC Mask Partical Filtration Efficiency Tester ( PFE ) with PSL Test Aerosol
T433S mask filtration efficiency tester is suitable for testing 45% -99.9% grade filter material efficiency test, and the filter material is tested according to international standards. The design structure of the whole tester is compact and light, and in the process of running the test, it only needs to provide additional suitable power and compressed air (no water, no oil), suitable for use in laboratories, factories and other occasions.
In the filter paper test, the particle counting method is used to measure the counting efficiency. It is recommended to use monodisperse aerosol in the test. The pressure drop measurement should be performed at the specified filtration rate.
Fix the filter paper on the fixture, and the test air volume corresponds to the required filter speed. The aerosol generated by the aerosol generator is adjusted, then electrostatically neutralized (PSL), and then evenly mixed with the filtered test air, and then enters the test area to penetrate the filter material.
In addition, the resistance of the filter material during the test must be measured and the flow rate adjusted manually.
GB 19082-2009 Technical requirements for medical disposable protective clothing
GB 19083-2010 Technical requirements for medical protective masks
GB / T 32610-2016 Technical Specifications of Daily Protective Masks
GB 2626-2006 Respiratory protective equipment-Self-suction filter anti-particulate respirator
YY 0969-2013 Disposable medical masks
YY 0469-2011 medical surgical mask
ISO-29463 high-efficiency filters and filter media for removing particulates from the air
|Mask Filtration Efficiency Tester|
|Test efficiency range||
45%-99.95%@≥ 0.3μ m
|Test pressure range||0-50pa ± 1%, 50-500pa ± 1.5%|
|Test flow of the tested filter material||
25-100L / min ± 2.5% (adjustable)
|Test aerosol type||
|Counter sampling flow rate||
2.83L / min ± 5%
|Particle size display||
|Counter laser light source||Laser diode|
|Laser Particle Counter||
Products comply with JJF1190-2008
|Imported air pump||
Works continuously and stably
|Data memory capacity||
1∽ 1000 sets of measurement data, exportable
Color 7-inch touch screen
|Printing method||built-in thermal printer|
|Power supply||AC 220V / 50Hz|
|Data export interface||USB|
0∽ 40 ℃ relative humidity 10∽ 70%
0 ℃ ∽ 45 ℃ Relative humidity 0∽ 70%
Thermal paper, hose (outer diameter 6mm and 15mm), instruction manual, test report, certificate of conformity
Length 1000X Width 760X Height 1750 (mm)
|Material||Shell steel plate paint, test parts SUS304 stainless steel|
|PSL aerosol generator|
1. Aerosol generator
It is mainly used for experiments requiring high-concentration aerosols. It uses 0.3um PSL standard particles. The size of the main particles emitted is ≥0.3um, accounting for more than 70% of the total. It can produce aerosols continuously and stably. Provide the test particles upstream of the filter material.
2. Particle Counter
It is mainly used to monitor the upstream and downstream particle concentration of the measured filter material to obtain the efficiency of the measured filter material.
3. Stainless steel fixture
It is mainly used for the compression and sealing of the tested filter material.
4. Differential pressure sensor
It is mainly used to test the pressure difference (or resistance) of the tested filter material under a certain test flow rate.
5. Electrical control and software
Equipment circuit control and operation, software calculation and data processing, upstream concentration stability judgment, operation control, parameter setting.
6. Imported air pump
Mainly used for the test flow through the filter material, the flow is stable, green and energy saving, and the vibration noise is small.
Contact Person: Emily