Payment & Shipping Terms:
|Product Name:||U - Shaped Pendulum Digital Charpy Metal Impact Testing Machine ( 150J / 300J ）||Impact Energy:||300J (large Hammer) 150 (small Hammer)|
|Impact Speed:||5.2m/s||Pendulum Pre-angle:||150 °|
|Sample Support Span:||40mm||Impact Blade Fillet:||R2.0~2.5mm R8mm|
|Angle Accuracy:||± 0.1 °||Key Words:||Charpy Metal Impact|
rubber tensile testing machine,
rubber testing equipments
U - Shaped Pendulum Digital Charpy Metal Impact Testing Machine ( 150J / 300J )
Touch screen control digital display impact testing machine is a new type of impact testing machine launched by our company in the near future. After continuous technical update and improvement in recent years, the product has reached the domestic advanced technology level, JB-S300B The digital pendulum impact tester is designed in strict accordance with the "GB/T3038-2002 pendulum impact tester" and GB/T229-2007 "metal Charpy notched impact test method", according to JJG145-82 "pendulum type The impact test machine is factory-examined, and the American standard impact cutting edge can be converted according to the actual needs of the user.
|Impact energy||300J (large hammer) 150 (small hammer)|
|Pendulum pre-angle||150 °|
|Sample support span||40mm|
|Support jaws rounded||R1.0~1.5mm|
|Impact blade fillet||R2.0~2.5mm R8mm|
|Distance from pendulum center to impact point||750mm|
|Standard sample size||10 (7.5, 5) × 10 × 55mm|
|Power and power||three-phase four-wire 50Hz 380V|
|Angle accuracy||± 0.1 °|
|Host size||2000 × 650 × 1960mm|
|Digital display cabinet size||360 × 400 × 1200mm|
|Surrounding environment.||There is no corrosive medium, no vibration and no strong electromagnetic field interference in the surrounding environment.|
Main equipment configuration:
Introduction to touch screen operation:
1. Bilingual interface:
Supports both Chinese and English operating languages, which can be switched between Chinese and English before entering the test interface;
Manually take the pendulum during the test. After the completion, the material is manually discharged, and then the unloading and impact are completed, and the device automatically takes the pendulum. The impact area and impact toughness will be displayed in the system display area after each experiment. The three tests are a group, and the system automatically calculates the average value of the three impact energies after the completion of one set.
3. Storage function:
It can store 100 sets of experimental data, including test name, test original value, average value, test time and other information.
Print function: (optional) Print the current test data through a micro printer.
4. Multi-level energy: The device supports multiple impact pendulums of different energy levels.
The local screen has a total of the boot language selection interface, the main operation interface, the memory recall interface, the password input interface, the system settings and the print screen, and each of the six in English and Chinese.
Test boot interface
1), Select Chinese and English system before entering the test , Chinese and English test interface
2),The main working interface of the device, implementing various actions of the device in this interface
3), Sample information input interface
This interface is the interface for inputting the basic information of the sample, preparing for calculating the impact toughness.
4), Test result print interface
After the test, press the print button in the sample information interface to display the preview of the printed report, and then press the print button in the print preview to print the test result. (The external micro-printer must be connected before printing. The printer should be configured by itself. The user must specify when purchasing is it a needle or a thermal printer?
Contact Person: Vilya