Payment & Shipping Terms:
|Applicable Industry::||Battery Industry||Observation Window::||Toughened Glass|
|Inner Container::||SUS304 Stainless Steel||Weight::||About 220kg|
|Sample Size (L * W * H)::||800*630*1780mm||Delay Function::||0~999min Could Be Set Arbitrarily|
|Outer Container::||Cold-roll Steel Sheet With Stoving Varnish||Warranty::||12 Months|
phone testing equipment,
mobile testing equipment
Battery Crush Squeezing Test Machine
GB31241-2014,IEC62133,UL1642,UL2054,QC/T743-2006,QC/T744-2006,GB8897.4-2002,IEC60086-4,SN/T1414,SN/T1413,YD1628,GJB2374,QB/T2052,IEC62133-2012,SAE-J2464,GB31241,YD/B 032-2009,YD/T 2344.1-2011,GB/T18332-2009,MT/T1051-2007
1.Clamp down through the vise or hydraulic pressure cylinder with cylindrical piston , make the single battery or battery unit be squeezed between the two planes. Continue to squeeze,don’t stop until the pressure reached about 13kN. Safety performance of the battery should be examined by squeezing test.There should be no fire breaking out, no explosion during the test. The testing equipment is also essential to the battery manufacturers and research laboratory.
2.For the simulation of single battery in process of use, storage, transportation, inspection when it has been penetrated or pierced through sharp objects,which cause the battery internal short circuit.Pierce the center position of the battery’s maximum surface with high temperature resistant steel needle,whose diameter is 2-8mm, at thespeed of 10mm/s-40mm/s, and keep it for any time. The battery is qualified by no explosion,no fire.
Specification and Parameter:
1.Power transmission mode: hydraulic pressure
A.the single battery is squeezed between the two parallel planes
B.the extrusion pressure of the plane is hydraulic pressure transmission
C.After the battery is firmly clamped, start to squeeze,don’t release until the pressure between two planes climb to 13kN.
Direction of squeezing:
For cylindrical cells, the long axis of the extruded cell should be parallel to the extrusion plane of the extrusion device,
For the rectangular battery, the squeezing pressure should be applied to the axial whose direction is perpendicular to one of the two axes of the battery.Squeeze the other axial next time.
For the button battery, squeeze the plane. Each test battery is only pressed once.
1,Door structure:The door of the box is provided with a double-layer visible explosion-proof glass, and the back of the box is provided with a pressure relief device and an exhaust tube, which is convenient for external connection of the laboratory large exhaust fan.
2,Observation window:The front of the door is opened with toughened glass window, the size is 300*300mm.
3,Delay function:0~999min could be set arbitrarily.
4,Material of the explosion-proof box:
Outer container :cold-roll steel sheet with stoving varnish
Inner container: SUS304 stainless steel
5,Weight: about 220kg
6,Power supply:380V, 50HZ
7,Machine size: 800×630×1780mm(L*W*H)
Contact Person: Cathy