Payment & Shipping Terms:
|Test Method:||Quality Act (MFR)||Temperature Range:||Room Temperature——450℃|
|Temperature Accuracy:||±0.2℃||Temperature Resolution:||0.1℃|
|Timing Range:||0.1——999 Seconds||Timing Accuracy:||0.1 Seconds|
|Product Name:||GB / T3682 Digital Plastic MFI Melt Flow Index Test Equipment For ABS Resin , Nylon||Key Words:||Melt Flow Rate|
rubber tensile testing machine,
rubber testing equipments
Plastic MFI Melt Flow Index Test Equipment is used to determine the melt flow rate of thermoplastics (MFR; units: g/10min) instrument. For polyethylene, polypropylene, polyoxymethylene, ABS resin, polycarbonate, nylon, plastics and other polymers fluoride melt flow rate at high temperature determination.
How it works:
Plastic MFI Melt Flow Index Test Equipment is in the specified temperature conditions, with high-temperature furnace to melt the state of the measured object.
The molten state of the measured object, the required weight load by gravity out of a certain diameter hole in the test.
Plastics in industrial enterprises and scientific research units of production, often with "melt (mass) flow rate" to represent the polymer material in the molten state of fluidity, viscosity and other physical properties.
Melt flow rate refers to the extrudate, the average quality of the sample paragraphs are translated into 10 minutes of extrusion volume.
Melt (mass) flow rate expressed by MFR, unit: g / 10 minutes (g / min) Formula: MFR (θ, mnom) = tref × m / t = 600 × m / t (g/10min) Where: θ - test temperature mnom - Nominal Load Kg
" m " - the average quality of cut g
" tref " - reference time (10min): 600
" t " - cut off time interval unit: seconds.
A set of plastic samples, a cut every 30 seconds, the quality of the results section: 0.081 g, 0.086 g,
0.081 grams, 0.089 grams, 0.082 grams.
Cut the average quality m = (0.081 +0.086 +0.081 +0.089 +0.082) ÷ 5 = 0.0838 ≈ 0.084 (g)
Into the formula: MFR = 600 × 0.084/30 = 1.680 (g / 10 min)
Plastic MFI Melt Flow Index Test Equipment for factories and enterprises and research institutes in the
production and research.
Meet the criteria:
Plastic MFI Melt Flow Index Test Equipment design and manufacture in accordance with the requirements
|Model||Plastic MFI Melt Flow Index Test Equipment|
|1. Squeeze out some material|
|Discharge port diameter||Φ2.095±0.005mm|
|Discharge port length||8.000±0.025mm|
|Loaded barrel diameter||Φ9.550±0.025mm|
|Loaded barrel length||152±0.1mm|
|Piston head diameter||9.475±0.015mm|
|Piston head length||6.350±0.100mm|
|2.Standard test load (eight grades)|
|1grade||0.325kg=(rod + weight + thermal units +1 No. tray body weight)|
|2grade||1.200 kg=(0.325+2 0.875 eight)|
|3grade||2.160 kg=(0.325+3 1.835 eight)|
|4grade||3.800 kg=(0.325+4 3.475 weight)|
|5grade||5.000 kg=(0.325+5 4.675 weight)|
|6grade||10.000 kg=(0.325+5 4.675 weight+6 5.000 weight)|
|7grade||12.500 kg=(0.325+5 4.675 weight+6 5.000+7 2.500 weight)|
|8grade||21.600 kg=(0.325+2 0.875+3 1.835+4|
|3.475+5 4.675+6 5.000+7 2.500+8 2.915 weight)|
|Relative error of Load Test≤0.5%|
|Test method||Quality Act (MFR)|
|Temperature range||Room temperature——450℃|
|Timing range||0.1——999 seconds|
|Timing Accuracy||0.1 seconds|
|Material cutting method||Automatic or manual way of cutting material|
|Equipped with a printer, can print test report|
|Working conditions||Ambient temperature10℃—40℃; relative humidity30%—80%|
|Dimensions||250*350*510(L* W* H)|
Contact Person: Vilya